1.It is to test voltage ratio, vector, polarity, ratio error of current transformer and voltage transformer.
2.The voltage ratio measurement for three or singe phase transformer
3.The group vector measurement for three phase transformer.
4.The polarity measurement for single phase transformer or CT/PT
5.Voltage ratio error calculation
6.Current transformer ratio measurement
7.Excitation current measurement in ratio testing
8. Reproduction the AC test voltage from power source to reduce the effects of bad power source
9.Save the test results in the memory internal the tester
10.Print the test results by micro printer
11.Reverse connection detect unit to prevent the high voltage injected to operator
12.Over current protection unit integrated in software and hardware
13.Upload the test results to PC by RS232 and make word report by software
Ratio measurement range:1~10000
Group measurement:1~12Beverage bottle, cosmetic bottle, medicine bottle, jar etc.
Ratio measurement error:1~2000 ± 0.2%RDG 2000~10000 ± 0.5%RDG
Power supply:AC220V ± 10%, 50Hz/60Hz
Size 400´350´200 mm3 Weight 7kg Humidity: ≤85%RH